Gradient (recalled) echo sequence
Spin echoes work fine for sequences of a long TR. If a short TR is needed (for example, in T1 weighted scans), we need to cut down the scan time. We do this by forgoing the 180° RF pulse and instead using a gradient to rephase the spins. This is a gradient echo sequence.
- RF pulse applied
- Slice-select gradient applied
- Phase-encoding gradient applied
- Frequency-encoding gradient applied
- A negative GFE is applied. The spins dephase, some faster than others.
- The positive GFE is applied. The spins start to rephase until they are again in phase and a signal is created - the Gradient Echo
The other aspect of a GRE sequence is that you don't have to use a 90° RF pulse at the start of the cycle, an RF pulse of any flip angle can be used. If an RF pulse with a smaller flip angle is used, it will take less time for the spins to regain all their Mz as they are closer to 0°. However, this also means that the Mxy is not as high as if a 90° flip angle was used.
Now available as an eBook to download and read anytime!
Witten by radiologists, for radiologists with over 220 beautiful diagrams optimised for Amazon Kindle. The proceeds of this book will help us to maintain the Radiology Cafe website.
Weighting using GRE sequences
|Flip angle||Small angle||Large angle||Small angle (minimise T1w)||Can't achieve T2w as no 180° RF pulses to cancer T2* effect|
|Spin echo||Gradient echo|
|RF pulse used to rephase||Gradient applied to rephase|
|Uses flip angle of 90°||Uses variable flip angle|
|Slow sequence||Fast sequence|
|True T2 weighting||T2* weighting - susceptible to magnetic field inhomogeneities|